TIANCHI used equine ultrasound machine for sale TC-210 Manufacturer in TV
Use of the probe
In order to prolong the probe's service life and obtain optimum performance, please follow these instructions:
1. Periodic inspection on probe cable, socket and acoustic window.
2. Shutdown the device before connecting or disconnecting the probe.
3. Do not drop the probe or flint body,and never hit the probe acoustic window, otherwise the probe will be damaged.
4. Do not attempt to heat the probe.
5. Do not attempt to bend or pull probe cable, otherwise the internal connection will be broken.
6. The Gel should be placed only on probe’s header and after using it, clean with a damp cloth.
7. Inspect the probe acoustical window, enclosure and cable carefully after probe cleaning. Never use the probe again if any crack or breakage is found.
Company:Henan Tianchi Instrument Equipment Co.,Ltd.
Matters needing attention?
1) According to the design, production and other related documents, the life expectancy of this model is generally 6 years, the product in excess of its life after the continued use, may produce performance degradation and failure rate significantly higher
2) In the normal and reasonable use, our products one year warranty host, probe warranty of two years, the battery warranty for six months. More than the warranty period or for non normal use (such as: accidentally broken water phenomenon) we only charge free maintenance, replacement of parts cost.
3) We set up a user file for each customer, product maintenance or replacement, just by the warranty card, and enjoy life-long technical services.
1) Service life:The service life of the device is 6 years.
(1)Discard the device according to local law.
(2)Do not discard it mixing with other household garbage.
The manufacturer shall not assume the responsibility of risks caused by using the device beyond its service life.
(1)Check the device power cord and probe cable and waterproof cover, if find any damages or breakages, do not use the device and replace the broken one immediately.
(2)Check whether the probe and main unit are connected rightly.
(3)Check the adapter EPS regularly, when the supply voltage exceeds specified accommodation limit (AC100V-240V±10%, 50/60Hz),do not supply the main unit or charge the battery with adapter.
(4)Check the adapter power cord and probe cable, if find any damages or breakages, replace the broken one immediately.
(5)Adapter is the dedicated power of the device, it adopts omniseal insulation design, do not replace it with other adapters or attempt to open it in case there be any hazard.