TIANCHI ultrasound medical equipment TC-301 Manufacturer in SG
Installation and disassembly
Connection between probe and main unit
Probe jack lies in the top of the right side, insert the probe into the jack correctly and turn it clockwise and wound it. There is only one probe jack which can be connected to all the supported optional probes.
Avoid by all means unplugging or plugging the probe connector at state of log on in case the probe and main unit be damaged. Once the probe is connected with the main unit, do not unplug nor plug it at discretion in case poor contact happen.
Must not touch the contact pin of the probe connector.
The probe should be protected from felling off or crashing and the manufacturer assumes no responsibility for this kind of hazard.
Please handle the device carefully.
Company:Henan Tianchi Instrument Equipment Co.,Ltd.
Probe:7.5MHz Linear Probe/6.5MHz Linear Rectal Probe/3.5MHz Convex Probe/5.0MHzMicro-convex Probe
veterinary autoclave :6.4 inch TFT-LCD
Image permanent storage:Image permanent storage
Gain range:0-192dB(adjust range 64-192dB)
Image process:Faked color,gray calibration,image smooth and histogram.
Frame correlation adjust:Support
Measuring functions:Distance,circumference,area,volume,EF rate
Time to empty:>3 hours
1) Operating environment
1. Environment temperature range: +10??+40?
2. Relative humidity range: 30%?75%
3. Atmosphere pressure range:70KPa?106KPa
While using, avoid strenuous vibration, keep it away from devices with high field, intense magnetic field or high voltage; avoid strong sunlight blazing down on the display; keep the device well-ventilated, moisture proof and dustproof.
2) Unpacking inspection
After unpacking, check the device according to "Packing List" and install it according to requirements and methods described in "Installation" after affirming that there is no shipping damage.
If there is breakage at unpacking check, it is banned to use the device to ensure security.
1. Install the battery correctly into the main unit.
2. Insert the plug of output DC14V/3A of the adapter into the DC14V interface on the side.
3. Connect power cord of "Input AC220V,50Hz" of the adapter to the EPS. <0.
4. No matter the main unit is power on or shut down, when the "under-voltage" indicator flickers, the adapter is charging the battery; When the "under-voltage" indicator turns off from flickering, the battery is fully charged.