Handheld optic power meter Fiber Optic Tester intelligent optical power meter together with laser source
3C-3208A handheld optic power meter can be used for the absolute power measurement and relative measurement of the link loss.
3C-3208A intergrades the handheld optical power meter and the intelligent optical power meter in one unit.
It can be used for the absolute power measurement and relative measurement of the link loss in dB.
Its compact size, friendly operation interface, broad power measurement range, high precision and brand-new
user automatic calibration function and high performance in application makes it an ideal tool for optical fiber network.
Handheld Power Meter Specification Model:
Handheld Power Meter A--Wavelength: 850nm, 980nm, 1300nm, 1310nm, 1490nm,1550nm,
Measurement Range: -70 to +3dBm
Handheld Power Meter C--Wavelength: 850nm,980nm , 1300nm, 1310nm, 1490nm,1550nm,
Measurement Range: -50 to +26dBm
- - - -
Model | Handheld Power Meter A | Handheld Power MeterC |
Wavelength Range(nm)*1 | 800~1600 |
Detector Type | InGaAs |
Measurement Range(dBm)*2 | -70~+3 | -50~+26 |
Uncertainty *3 | 0.15dB |
Accuracy | Linear 0.1% or Nonlinear 0.01dBm |
Operating Temperature() | -10~+60 |
Storage Temperature() | -25~+70 |
Auto-off Time(min) | 10 |
Operating Time(h) | >=130 |
Battery | 3 AA1.5V |
Weight(g) | 200 |
Dimensions(mm) | 152*74*26 |
- - - -
1. Wavelength Range: Specified standard operating wavelength range in which the Power Meter can work properly under certain technical specifications.
2. Power Measurement Range: The maximum and minimum range in which the Power Meter can work properly.
3. Uncertainty: Difference between two measurement results that were tested by Power Meter and another Standard Power Meter respectively.
1. It is important to keep all optical connectors and surfaces free from oil, dirt or other contamination to ensure proper operation.
2. Keep using one type of adapter to avoid excess loss from different connectors.
3. Please use dust proof cap for protection to avoid begin scratched or contaminated when Handheld Power Meter not in operation.
4. Light interface is sensitive, please carefully plug in and pull out the connectors.
5. Please use clean cotton to clean the sensor surface, clean it in clockwise direction carefully.
6. If does not need to use for a long time, please take out the battery.
1. Company information
Main product: SFP,Media Converter,Video Transmitter,Passive Product,Passive Equipment, Active Product and so on.
Location: Guangdong, China(Mainland)
Year Established: 2008
Main Markets: Eastern Europe, North America, Oceania, Africa, Southeast Asia
Certifications: ISO Certificate
Product Certifications: CE,RoHS
2. Contact us
Contact person: Linn Tan